Frequency Instability Measurement Device Based on the Pulse Coincidence Principle
Keywords:short-time frequency instability, stand-alone measurement device, virtual measurement device, LabVIEW, packet coincidence of pulses
Context. The task of rapid and accurate measurement of the dynamic characteristics of modern signal sources with a frequency output, in particular, the short-time frequency instability function, calls for refining measurement techniques with account of the requirement to improve their metrological characteristics, reduce test time, and automate measurements by using information-and-measurement systems.
Objective. The goal of the work is to develop a method of measuring the short-time frequency instability function using the principle of pulse packet coincidence and experimental investigation of measurement devices based on this principle.
Method. A method was developed for measuring the short-time frequency instability function based on the principle of packet coincidence of regular independent pulse trains. The developed method has advantages over the best version of the method based on the period-time interval-code (PTC) conversion when working with the same initial value of the investigated frequency and when working with the same value of the averaging interval.
Results. Analytical relationships were obtained for basic metrological characteristics. A comparative analysis was carried out for the metrological characteristics of the developed method and the method using period-time interval-code conversion. Acceptable metrological characteristics are inherent to the short-time frequency instability function (SFIF) measurement method based on the period-time interval-code technique. The difference of investigated and reference intervals form the measurement interval, which is filled with pulses of the investigated or reference frequencies.
Conclusions. Stand-alone and virtual measurement devices were developed, and experimental studies of standard oscillators were carried out. The features of measurement devices were specified and the ways of their further improvement were described. Further development of the measurement device can involve an increase in the number of measured signal source with frequency output (SFO) parameters, in particular, changes in short-time frequency instability due to the action of destabilizing factors, and the characteristics and time of frequency setting. This calls for developing a controlled source of destabilizing factors and synchronizing its operation with the measurement device.
Dawkins S., McFerran J. and Luiten A. (2007) Considerations on the measurement of the stability of oscillators with frequency counters. IEEE Transactions on Ultrasonics, Ferroelectrics and Frequency Control, Vol. 54, Iss. 5, pp. 918-925. DOI: 10.1109/TUFFC.2007.337
Bourgeois P., Kersale Y., Bazin N., Chaubet M. and Giordano V. (2004) Measurement of short-term stability of ultra stable oscillators. 18th European Frequency and Time Forum (EFTF 2004). DOI: 10.1049/cp:20040920
Kychak V. and Gavrasienko P. (2014) Device of Radio Frequency Instability Measurements. Visnyk NTUU KPI Seriia - Radiotekhnika Radioaparatobuduvannia, No. 58, pp. 83-89. DOI: 10.20535/RADAP.2014.58.83-89.
Nsue J.M.B., Kucheryavenko S.V. and Fedosov V.P. (2018) Measurement of short-term instability of the frequency of superstable quasi-hormonic signals. Engineering Journal of Don, No. 1, 8 p.
Milanovic I., Renovica S., Župunski I., Banović M. and Rakonjac P. (2012) How To Measure Oscillator’s Short-Term Stability Using Frequency Counter. Electronics ETF, Vol. 16, Iss. 1. DOI: 10.7251/els1216104m
Keysight Technologies (2014) Analyzing Frequency Stability in the Frequency and Time Domains, Application Note, p. 12.
Goryaschenko K. L., Goryaschenko S. L., Gula I. V. and Trotsyshyn I. V. (2016) Method for measuring phase shift between periodic signals of arbitrary duration. Patent UA111253.
Verveyko A. I., Tyrsa V. Ye., Uvarov V. M. and Shmalii Yu. S. (1987) Device for measuring frequency instability. Patent SU1442928
Verveyko A. I. (2013) Virtual device for measurement of the short-time frequency instability function. Bulletin of the Chernihiv State Technological University, No. 4 (69), pp. 105-115.
Tyrsa V. Ye. (1981) Analysis of frequency measurement errors based on the pulse coincidence principle. Izmeritelnaya tekhnika, No. 4, pp. 42-44.
Verveyko A. I., Yevdokimenko Yu. I., Tyrsa V. Ye., Shmalii Yu. S. (1989) Device for measurement of dynamic characteristics. Patent SU1532901
Pavlovsky V., Verveyko A., Shkola I. (2003) Reconfigured laboratory complexes in educational process of the Chernihiv State Technological University. Advanced Computer Systems and Networks: Design and Application (ACSN-2003), Lviv, pp. 46-47.
Jerone J. (2010) Virtual instrumentation using LabVIEW. New Delhi, PHI Learning Private Limited, 446 p.
Guk M. Yu. and Zaborovsky V. S. (2008) Telematics applainces architecture for active experiments with remote objects. SPbPU Journal. Computer Science. Telecommunication and Control Systems, No. 2 (55), pp. 46-61.
Bress T. (2013) Effective Labview Programming. USA, Austin, NFS Press, 816 p.
IJ J.E. and Loureiro O.S. (2015) Research and Development of a Virtual Instrument for Measurement, Analysis and Monitoring of the Power Quality. Journal of Fundamentals of Renewable Energy and Applications, Vol. 05, Iss. 05. DOI: 10.4172/2090-4541.1000185
Guo H. and Wu J. (2014) Application of Virtual Instrument LabVIEW in Variable Frequency and Speed Motor System. TELKOMNIKA Indonesian Journal of Electrical Engineering, Vol. 12, Iss. 8. DOI: 10.11591/telkomnika.v12i8.5571
How to Cite
Authors who publish with this journal agree to the following terms:
- Authors retain copyright and grant the journal right of first publication with the work simultaneously licensed under a Creative Commons Attribution License that allows others to share the work with an acknowledgement of the work's authorship and initial publication in this journal.
- Authors are able to enter into separate, additional contractual arrangements for the non-exclusive distribution of the journal's published version of the work (e.g., post it to an institutional repository or publish it in a book), with an acknowledgement of its initial publication in this journal.
- Authors are permitted and encouraged to post their work online (e.g., in institutional repositories or on their website) prior to and during the submission process, as it can lead to productive exchanges, as well as earlier and greater citation of published work (See The Effect of Open Access).